XXQ-2505 Portable directional X ray detector with directional
|Output voltage||150-250 KV|
|Focus size1.5||2.0×2.0 mm|
|Penetration||FE 39 mm|
|Generator size||320×320×750 mm|
|Controller size||380×380×190 mm|
|Controller weight||12 kg|
|Input power||AC 185-245V @50~60Hz|
1. It can still work by applying special design what can avoid the
interference by power grid change.
2. We are using highpower electronic components,such as capacitance
and power module with wide capability to greatly ensure the quality
of the products.
3. Apply the isolation technology of low voltage and high voltage
to ensure the controller working stablely.
4. Posses the reliable “time-delayed” function, suitable to operate
in the open field.
5. It has functions of troubleshooting automatically and pressure
indication to guarantee the stability of working.
-- Compact x-ray generator, anode grounded, forced air cooling
-- Light weight, portable, easy to use
-- The ratio of the exposure fine to the stop time is 1:1
-- High voltage ,time delay function protects the operator from
-- Strictly controlled the production techniques, nice sculpt,
Composition and Application
x ray flaw detector consists of three parts,controller, generator
and connection cable .
X Ray flaw detector plays important parts in carrying on
nondestructive examination for nonmetallic, light metal, casting,
various alloys and pressure vessels etc, mainly detecting weld
defects ,such as crack, blowhole, entrapped slag, incomplete fusion
,blowhole,bubble etc ,also using for shipbuilding ,airplane and
national defense etc industry fiel.
Matters needing attention
(1) After power on, the controller has a high internal AC and DC
voltage, the use of the case is not allowed to open the case, non
maintenance personnel do not repair.
(2) When not in use, the equipment should be placed in a dry and
ventilated place, the temperature is suitable for a long time did
not use the equipment, when enabled to see if wet and other
phenomena, if there is to be treated before use.
(3) The temperature of the container X ray generator, with the use
of time may be rising, handling equipment on cameras, please wear
gloves handling, and must be handled with care.
(4) In handling X mobile X-ray machine, if severe vibration or
shock, can make X ray tube fragmentation or filament deformation or
high voltage coil move, resulting in equipment is not working
properly or damaged.
(5) The X ray generator in the car, forklift, must stand (fan end
in), while below the mat cushion, if must be sideways when carrying
rings at both ends to pad the damping material to prevent shock.