XXG-2005A Smaller sized Industrial X ray Machine with Generator
weight 19kg applications Penetration 30mm
|Output voltage||100-200 KV|
|Focus size||1.5×1.5 mm|
|Generator size||230×230×580 mm|
|Controller size||310×210×130 mm|
|Controller weight||7 kg|
|Input power||AC 185-245V @50~60Hz|
1. It can still work by applying special design what can avoid the
interference by power grid change.
2. We are using highpower electronic components,such as capacitance
and power module with wide capability to greatly ensure the quality
of the products.
3. Apply the isolation technology of low voltage and high voltage
to ensure the controller working stablely.
4. Posses the reliable “time-delayed” function, suitable to operate
in the open field.
5. It has functions of troubleshooting automatically and pressure
indication to guarantee the stability of working.
-- Compact x-ray generator, anode grounded, forced air cooling
-- Light weight, portable, easy to use
-- The ratio of the exposure fine to the stop time is 1:1
-- High voltage ,time delay function protects the operator from
-- Strictly controlled the production techniques, nice sculpt,
Composition and Application
x ray flaw detector consists of three parts,controller, generator
and connection cable .
X Ray flaw detector plays important parts in carrying on
nondestructive examination for nonmetallic, light metal, casting,
various alloys and pressure vessels etc, mainly detecting weld
defects ,such as crack, blowhole, entrapped slag, incomplete fusion
,blowhole,bubble etc ,also using for shipbuilding ,airplane and
national defense etc industry fiel.
Matters needing attention
(1) Because of the system work, to produce a high voltage pulse
controller, induction to the housing, to connect the ground when
necessary. When erecting generator in welding plate and the
workpiece, the generator with iron and workpiece distance, avoid
the welding current through the generator, the low voltage cable
into the control box will damage to the machine.
(2) X ray generator and controller are forced air cooling, the fan
will cool air into the device, so the device is not in use, sand
dust, moisture and other occasions. Attention should be paid to the
vents, exhaust port should not be obstructed, if the cooling fan is
stopped in a short period of time, the temperature will rise
failure, so every day before use to check.
(3) The input AC voltage at 220V + 10% range, if the input voltage
is too high (such as 380V) will use the instrument to burn out. If
the input voltage is too low, the instrument can not work normally.